|Year : 2020 | Volume
| Issue : 2 | Page : 127-134
Comparative evaluation of the precision of cone-beam computed tomography and surgical intervention in the determination of periodontal bone defects: A clinicoradiographic study
Unnati Pitale1, Hitesh Mankad1, Rohit Pandey1, Pritish Chandra Pal1, Shikha Dhakad2, Antush Mittal1
1 Department of Periodontics, Modern Dental College and Research Centre, Indore, Madhya Pradesh, India
2 Department of Periodontics, Chhattisgarh Dental College and Research Institute, Rajnandgaon, Chhattisgarh, India
|Date of Submission||25-Feb-2019|
|Date of Decision||25-Oct-2019|
|Date of Acceptance||22-Dec-2019|
|Date of Web Publication||02-Mar-2020|
Dr. Unnati Pitale
Department of Periodontics, Modern Dental College and Research Centre, College Campus, Staff Quarter - 202, Block A, Airport Road, Gandhinagar, Indore - 453 112, Madhya Pradesh
Source of Support: None, Conflict of Interest: None
| Abstract|| |
Context: Major limitations of conventional radiography are overlapping and lack of 3D information. Surgical exposure, though being able to provide accurate information, provides very little time to plan-out the type of periodontal regeneration required during surgery. Cone Beam Computed Tomography (CBCT) has emerged as a feasible tool and found to be accurate. Unfortunately, in-vivo studies are still scarce. Aim: Aim of the present study was to assess the efficacy of CBCT in the detection of periodontal bony defects while determining its quantitative precision in the measurement of alveolar bone height as against the open flap debridement (OFD) procedure which is set as the gold standard. Setting and Design: Present study is a cross-sectional study. Materials and Methods: The present study includes patients with Chronic Periodontitis indicated for periodontal surgeries. Bone defects were measured with the help of CBCT and with William's periodontal probe during surgical intervention and compared. Statistical Analysis: Measurements were compared with Student's t-test; unpaired t-test & correlation were tested with Pearson's correlation coefficient test. P < 0.05 was considered statistically significant. Results: The mean CBCT & surgical value of palatal/lingual & distal sites of anterior teeth showed statistically significant difference (P = 0.001). All the values for posterior teeth were statistically non-significant. Conclusion: Statistically CBCT & clinical measurement with OFD have similar potential of accuracy to access the bony topography but CBCT provides good accessibility to visualize the sites which are difficult to access during surgical interventions like palatal sites & the distal sites of the posterior teeth.
Keywords: Alveolar bone height, chronic periodontitis, cone-beam computed tomography, periodontal bone defects, UNC 15 periodontal probe
|How to cite this article:|
Pitale U, Mankad H, Pandey R, Pal PC, Dhakad S, Mittal A. Comparative evaluation of the precision of cone-beam computed tomography and surgical intervention in the determination of periodontal bone defects: A clinicoradiographic study. J Indian Soc Periodontol 2020;24:127-34
|How to cite this URL:|
Pitale U, Mankad H, Pandey R, Pal PC, Dhakad S, Mittal A. Comparative evaluation of the precision of cone-beam computed tomography and surgical intervention in the determination of periodontal bone defects: A clinicoradiographic study. J Indian Soc Periodontol [serial online] 2020 [cited 2022 Aug 19];24:127-34. Available from: https://www.jisponline.com/text.asp?2020/24/2/127/279591
| Introduction|| |
periodontal disease causes attachment loss, alveolar bone loss, and consequent tooth mobility., Radiographs play an important role in diagnosis because they can reveal the amount and type of damage. One major limitation of conventional radiography is the presence of considerable overlapping of the anatomical structures and lack of three-dimensional information.,, In a plethora of situations, this becomes an actual hindrance in distinction between the buccal and palatal/lingual cortical aspects, making it difficult to obtain an effective evaluation, especially the crater defects and furcation involvements.,,, This is most commonly observed when periapical, interproximal, and panoramic radiographs are used to evaluate the alveolar bone architecture. Thus, precision of the defect or hidden pathology cannot be ascertained, for which surgical exposure has been the only means to accurately detect the extent of the bone loss, as well as to evaluate the gain in bone height after a definitive periodontal treatment. Surgical exposure, though provides accurate information of the area, gives very little time to plan out for periodontal regeneration required. Recently, cone-beam computed tomography (CBCT) has emerged as a feasible tool in dentistry, providing a lower cost alternative to the conventional computed tomography (CT) with high-quality images and lower radiation exposure to the patients., For the detection of smallest periodontal osseous defects, CBCT can display the image in all its three dimensions by removing the interfering anatomical structures and making it possible to evaluate each root and surrounding bone.,,,,, CBCT is found to be as accurate as direct measurements using a periodontal probe and as reliable as intraoral periapical radiograph for the detection of interproximal areas, which are considered to be the highest of the resolution till date. Recent in vitro studies have shown better precision in the evaluation of bony changes associated with periodontal disease using CBCT when compared to the conventional and digital radiography,, but there are scarcity of in vivo studies. Therefore, the aim of the present study was to assess the efficacy of CBCT in detection of periodontal bony defects in chronic periodontitis cases and compare it with the open flap debridement procedure which is the gold standard for the evaluation of bony topography.
| Materials and Methods|| |
This study was designed as a cross-sectional study and patients were selected from the outpatient department. In inclusion criteria, the individuals between 25 and 55 years of age with moderate-to-severe chronic periodontitis which were indicated for surgery and with no history of systemic disease and tobacco use were included. All the individuals were clinically and radiographically evaluated as a routine diagnostic procedure with the exclusion criteria of pregnant and lactating females. In the present study, 12 patients with 12 teeth (6 anterior and 6 posterior) each were selected, and thus, a total of 144 teeth with chronic periodontitis were selected. For anterior teeth, 12 patients were considered with six teeth and eight sites as shown in [Figure 1]. The number of sites calculated was 12 patients × 6 teeth × 8 sites = 576. For posterior teeth, 12 patients were considered with six teeth and 12 sites as shown in [Figure 2]. The number of sites calculated was 12 patients × 6 teeth × 12 sites = 864. Hence, the total sites calculated as: anterior sites (576) + posterior sites (864) = 1440, which are shown in [Table 1] and [Table 2]. Presurgically, each patient received an initial full-mouth scaling and root planing, oral hygiene, and plaque control instructions. Once the patients showed plaque proficiency up to the mark, then surgeries were initiated. CBCT images were acquired for the same area of interest in a single 360° rotation around the head of the patient. The standard image-acquisition time was 36 s in which the patient received pulsed radiation. All images were then stored in Digital Imaging Communications in Medicine file format. Following image acquisition, measurements were made using the proprietary NNT software (Version 5.4, cefla dental group, newtom GiANO, Italy) integrated with Newtom-Giano CBCT equipment which has multiple adjustable field of views. The data were acquired with an image size of 400 × 400 × 400, voltage of 90 kV, voxel size of 400 μm, and current of 10 mA. The examiner underwent training for performing the measurements. The CBCT measurements were performed with the same pattern as clinical measurements of sites. In case of horizontal defects, distance between cementoenamel junction (CEJ) and alveolar crest was measured, and in case of vertical defects, the distance between CEJ and the base of the defect was measured. The viewer software accuracy was 0.2 mm. Informed consent was signed by every subject who fulfilled the inclusion criteria and agreed to participate voluntarily, after thorough explanation of the nature, risk, and benefit of the clinical investigations and associated procedure. The University Ethical Committee approved the consent form and experimental protocol (IEC No-IEC/MDCRC/2013–14/0106).
During the surgical intervention, periodontal osseous defects were measured with the help of UNC 15 periodontal probe to the nearest millimeter from the CEJ to the deepest level of the bony defect, which is considered as the “gold standard” in literature., Assessment of depth of the bony defect was measured during the surgical procedure in the same region where CBCT measurements were evaluated [Figure 3]a, [Figure 3]b and [Figure 4]a [Figure 4]b [Figure 4]c [Figure 4]d.
|Figure 3: (a) Measurement of bony defects around the mandibular molar teeth with a UNC 15 periodontal probe with surgical intervention; (b) Cone-beam computed tomography images of bony defects around the mandibular molar teeth in coronal, sagittal, axial plane and three-dimensional reconstruction of the area of interest|
Click here to view
|Figure 4: (a and d) Measurement of bony defects around the mandibular canine teeth with a UNC 15 periodontal probe with surgical intervention; (b) Cone-beam computed tomography images of bony defects around the mandibular canine in coronal, sagittal, axial plane and three-dimensional reconstruction of the area of interest; (c) Cone-beam computed tomography images of bony defects around the mandibular canine|
Click here to view
All the patients were instructed to rinse with 10 ml 0.2% chlorhexidine twice daily till 2 weeks postoperatively. Periodontal dressing and sutures were removed postoperatively after 7 days.
Clinical and CBCT measurements were compared in each anterior and posterior tooth with the help of Student's paired t-test. If P ≥ 0.05, the difference observed was considered statistically insignificant, and if P ≤ 0.05, it was considered significant. Pearson's correlation coefficient (z) test was used to test correlation. Unpaired t-test was used to compare the data in the same measurement method.
| Results|| |
All 144 teeth sites were measured and the mean of each site was calculated which comes as four sites (mean of 8 sites) in anterior and four sites (mean of 12 sites) in posterior as shown in [Table 1] and [Table 2] which is according to [Figure 1] and [Figure 2].
Descriptive values for anterior teeth
- Buccal sites – The mean CBCT value was 4.156 mm and mean surgical value was 4.195 mm. The mean difference was statistically nonsignificant (P = 0.171) [Table 3], [Table 4], [Table 5] and [Graph 1]
- Palatal/lingual sites – The mean CBCT value was 4.0444 mm and mean surgical value was 4.1822 mm. The mean difference was statistically significant (P = 0.001) [Table 3], [Table 4] and [Graph 2]
- Mesial sites – The mean CBCT value was 4.1547 mm and mean surgical value was 4.1982 mm. The mean difference was statistically nonsignificant (P = 0.775) [Table 3], [Table 4] and [Graph 3]
- Distal sites – The mean CBCT value was 3.3667 mm and mean surgical value was 3.5217 mm. The mean difference was statistically significant (P = 0.001) [Table 3], [Table 4] and [Graph 4]. All the measurements were found to be in correlation with each other.
|Table 3: Mean and standard deviation values of anterior teeth measured through cone-beam computed tomography and surgical intervention|
Click here to view
|Table 4: The comparison of t values for the anterior teeth by Student's paired t-test|
Click here to view
|Table 5: Pearson's correlation test for all sites of anterior teeth and the significance status of sites|
Click here to view
Descriptive values for posterior teeth
- Buccal sites – The mean CBCT value was 6.0656 mm and mean surgical value was 5.9833 mm. The mean difference was statistically nonsignificant (P = 0.150) [Table 6], [Table 7], [Table 8] and [Graph 5]
- Palatal/lingual sites – The mean CBCT value was 6.1111 mm and mean surgical value was 6.1556 mm. The mean difference was statistically nonsignificant (P = 0.405) [Table 6], [Table 7] and [Graph 6]
- Mesial sites – The mean CBCT value was 5.6956 mm and mean surgical value was 5.6444 mm. The mean difference was statistically nonsignificant (P = 0.192) [Table 6], [Table 7]and [Graph 7]
- Distal sites – The mean CBCT value was 6.0867 mm and mean surgical value was 6.0389 mm. The mean difference was statistically nonsignificant (P = 0.731) [Table 6], [Table 7] and [Graph 8].
|Table 6: Mean and standard deviation values of posterior teeth through cone-beam computed tomography and surgical intervention|
Click here to view
|Table 7: The t values for the posterior teeth by Student's paired t-test and significance status of sites by P value|
Click here to view
|Table 8: Pearson correlation test for all sites of posterior teeth and the significance status of sites|
Click here to view
- CBCT measurements – The mean difference for buccal and palatal/lingual sites was 0.446 mm which was nonsignificant. The mean difference for mesial and distal sites was 0.829 mm which was nonsignificant (P > 0.05) [Table 9]
- Surgical measurement – The mean difference for buccal and palatal/lingual sites was 0.444 mm which was nonsignificant. The mean difference for mesial and distal sites was 0.723 mm which was nonsignificant (P > 0.05).
|Table 9: Unpaired t-test for the surgical and cone-beam computed tomography measurements of all teeth and its significance status|
Click here to view
- CBCT measurement – The mean difference for buccal and palatal/lingual sites was 0.806 mm which was nonsignificant. The mean difference for mesial and distal sites was 0.446 mm which was nonsignificant (P > 0.05) [Table 9]
- Surgical measurement – The mean difference for buccal and palatal/lingual teeth was 0.360 mm which was nonsignificant. The mean difference for mesial and distal sites was 0.043 mm which was significant (P ≤ 0.05).
| Discussion|| |
CBCT has opened a new vista in maxillofacial imaging facilitating transition of diagnostics from two-dimensional to three-dimensional imaging and having the potential to expand the role of dental imaging from mere diagnosis to image guidance of surgical procedures. Numerous studies have validated the use of CBCT for planning of implants, in diagnosis of embedded teeth, in orthodontics and numerous pathologies;,,,,,,,,,,,,,,,,,,, however, very few have analyzed the need of CBCT in periodontal diagnosis, regeneration, and treatment planning. Recently, in vivo studies have been published concerning periodontal bone height measurements using CBCT. This leaves only a few reports on the accuracy of CBCT in measuring periodontal defects while almost none of them concerning horizontal bone defects.,,, The main advantages of CBCT include good accessibility and easy handling in addition to a real-size data set., Although Vandenberghe et al. suggested CBCT to be used only for relatively complex periodontal treatment planning and potential use of dental implants, this imaging modality may provide a new tool for diagnosis of various bony defects, interdisciplinary approach to treatment planning such as in case of periodontal–endodontic lesions and in the detection of dehiscence and fenestration defects before initializing the orthodontic treatments.
This present study had given us the clear idea about the precision of CBCT measurement and its status. The analysis of anterior teeth concluded that the mean difference for both the methods made for mesial and buccal surfaces was more precise in surgical intervention than CBCT analysis [Table 3], but it was statistically nonsignificant with P = 0.171 and 0.775, respectively. For palatal/lingual surface (P = 0.001) and distal surface (P = 0.001), the mean difference for both the methods was found statistically significant [Table 4]. The said results were found to be correlated with the results obtained in the study conducted by Mol and Balasundaram, who found less accuracy in the measurement of buccal bone defects in the anterior region of mandible in comparison with the posterior region on images with the New Tom 9000 Scanner. Furthermore, using helical CT, Fuhrmann et al. found that bone defects with wall thickness <0.2 mm were not apparent on tomographic imaging.
For posterior teeth, the mean difference of the buccal surface (P = 0.150) and palatal/lingual surface (P = 0.405) was statistically nonsignificant. Mesial and distal surface (P = 0.192 and P = 0.731) was also found statistically nonsignificant. The results obtained were in concordance with the results of the previous studies conducted by de Faria et al., Grimard et al., and Feijo et al. The difference in the diagnostic accuracy of CBCT in the anterior and posterior teeth is likely the result of the difference in the morphology of the periodontal bone in each region. The lingual plates are considerably thinner in the anterior region and tapers toward the alveolar crest than in the posterior teeth. A thinner bone plate has low image resolution decreasing the precision of linear measurements in case of anterior teeth. Similar to the present study, Misch et al. showed that CBCT analysis was as accurate as direct measurement using a periodontal probe during surgical intervention. However, within same analytical parameters, when unpaired t-test was applied, it was found that there was a significant difference between the mesial and distal sites in posterior teeth in the measurements made during surgical intervention. This difference can be explained by the fact that the distal site accessibility is difficult while performing periodontal surgeries, especially in case of molars while instrumentation as well as for measurements attempted with the help of standard periodontal probing. On the contrary, CBCT provides good accessibility to visualize the sites which are difficult to access during surgical interventions, clinically including the distal sites of posterior teeth. Similarly, Feijo et al. showed that a clinical measurement gives accuracy up to 1 mm whereas CBCT measurements allowed an accuracy of up to three decimal places. Ideally, 0.5 mm discrepancy between clinical and radiographical estimated bone levels is accepted. Smaller or larger errors in locating the CEJ and crestal levels can, respectively, lead to over- and under-estimation of disease. Because of the limitations of surgical re-entry to evaluate regenerative outcomes, different radiographic techniques such as intraoral radiographs, digital subtraction radiography, and computer-assisted densitometric image analysis have been used as surrogate methods. However, these techniques are difficult to standardize from two different time points and having limitation to assess three-dimensional morphologic changes. Hence, this present study gave us a view that the results of surgical intervention and CBCT give similar value and prove the accuracy and reliability of CBCT analysis. Data from CBCT could be helpful in optimization of surgical treatment plan. Involvement of furcation in case of posteriors can be determined accurately with CBCT. In nutshell, CBCT provides better evaluation of the bony topography, which gives precision in the type and depth of the defect pre- and post-operatively after regeneration. Surgical exposure, though being able to provide accurate information of the area, provides very little time for the surgeon to plan the treatment protocol for suitable regeneration required during surgery.
| Conclusion|| |
This present study compares CBCT analysis and surgical intervention method and concluded having similar potential of accuracy to access the bony topography in both the measurement methods. However, the diagnostic accuracy of CBCT is lesser for anterior teeth as compared to the posterior teeth due to thinner buccal/lingual cortical plates. The quality of the CBCT image slices is insufficient to resolve the alveolar crest reliably in the anterior region. However, CBCT having better accessibility and precision to visualize any area of interest, which is difficult to access during surgical intervention like distal surface of posterior teeth. The bone density/volume can be detected with supremacy by CBCT. It gives benefit to plan the treatment for better prognosis and its outcome, by avoiding surgical re-entry to evaluate postoperative regeneration precisely and help to improve futuristic approach.
Financial support and sponsorship
Conflicts of interest
There are no conflicts of interest.
| References|| |
Newman MG, Takei HH, Klokkevold PR, Carranza FA. Carranza's Clinical Periodontology. 11th
ed. St. Louis, MO: Elsevier/Saunders; c2002. p. 454-68.
Highfield J. Diagnosis and classification of periodontal disease. Aust Dent J 2009;54 Suppl 1:S11-26.
Pepelassi EA, Diamanti-Kipioti A. Selection of the most accurate method of conventional radiography for the assessment of periodontal osseous destruction. J Clin Periodontol 1997;24:557-67.
Cotti E, Campisi G. Advanced radiographic techniques for the detection of lesions in bone. Endod Topics 2004;7:52-72.
Mol A. Imaging methods in periodontology. Periodontol 2000 2004;34:34-48.
Eickholz P, Hausmann E. Accuracy of radiographic assessment of interproximal bone loss in intrabony defects using linear measurements. Eur J Oral Sci 2000;108:70-3.
de Faria Vasconcelos K, Evangelista KM, Rodrigues CD, Estrela C, de Sousa TO, Silva MA. Detection of periodontal bone loss using cone beam CT and intraoral radiography. Dentomaxillofac Radiol 2012;41:64-9.
Fuhrmann RA, Bücker A, Diedrich PR. Furcation involvement: Comparison of dental radiographs and HR-CT-slices in human specimens. J Periodontal Res 1997;32:409-18.
Eskandarlo A, Bardal R, Dehghani M. Accuracy of cone beam computed tomography intraoral radiography and periodontal probing for periodontal bone defects measurement. J Dent Med Tehran Univ Med Sci 2011;24:15-9.
Grimard BA, Hoidal MJ, Mills MP, Mellonig JT, Nummikoski PV, Mealey BL. Comparison of clinical, periapical radiograph, and cone-beam volume tomography measurement techniques for assessing bone level changes following regenerative periodontal therapy. J Periodontol 2009;80:48-55.
Scarfe WC, Farman AG, Sukovic P. Clinical applications of cone-beam computed tomography in dental practice. J Can Dent Assoc 2006;72:75-80.
Hounsfield GN. Computerized transverse axial scanning (tomography). 1. Description of system. Br J Radiol 1973;46:1016-22.
Banodkar AB, Gaikwad RP, Gunjikar TU, Lobo TA. Evaluation of accuracy of cone beam computed tomography for measurement of periodontal defects: A clinical study. J Indian Soc Periodontol 2015;19:285-9.
] [Full text]
Mohan R, Singh A, Gundappa M. Three-dimensional imaging in periodontal diagnosis – Utilization of cone beam computed tomography. J Indian Soc Periodontol 2011;15:11-7.
] [Full text]
Leung CC, Palomo L, Griffith R, Hans MG. Accuracy and reliability of cone-beam computed tomography for measuring alveolar bone height and detecting bony dehiscences and fenestrations. Am J Orthod Dentofacial Orthop 2010;137:S109-19.
Misch KA, Yi ES, Sarment DP. Accuracy of cone beam computed tomography for periodontal defect measurements. J Periodontol 2006;77:1261-6.
Pinsky HM, Dyda S, Pinsky RW, Misch KA, Sarment DP. Accuracy of three-dimensional measurements using cone-beam CT. Dentomaxillofac Radiol 2006;35:410-6.
Feijo CV, Lucena JG, Kurita LM, Pereira SL. Evaluation of cone beam computed tomography in the detection of horizontal periodontal bone defects: An in vivo
study. Int J Periodontics Restorative Dent 2012;32:e162-8.
Vandenberghe B, Jacobs R, Yang J. Diagnostic validity (or acuity) of 2D CCD versus 3D CBCT-images for assessing periodontal breakdown. Oral Surg Oral Med Oral Pathol Oral Radiol Endod 2007;104:395-401.
du Bois AH, Kardachi B, Bartold PM. Is there a role for the use of volumetric cone beam computed tomography in periodontics? Aust Dent J 2012;57 Suppl 1:103-8.
Mischkowski RA, Pulsfort R, Ritter L, Neugebauer J, Brochhagen HG, Keeve E, et al
. Geometric accuracy of a newly developed cone-beam device for maxillofacial imaging. Oral Surg Oral Med Oral Pathol Oral Radiol Endod 2007;104:551-9.
Tyndall DA, Rathore S. Cone-beam CT diagnostic applications: Caries, periodontal bone assessment, and endodontic applications. Dent Clin North Am 2008;52:825-41, vii.
Veyre-Goulet S, Fortin T, Thierry A. Accuracy of linear measurement provided by cone beam computed tomography to assess bone quantity in the posterior maxilla: A human cadaver study. Clin Implant Dent Relat Res 2008;10:226-30.
Januário AL, Barriviera M, Duarte WR. Soft tissue cone-beam computed tomography: A novel method for the measurement of gingival tissue and the dimensions of the dentogingival unit. J Esthet Restor Dent 2008;20:366-73.
Barriviera M, Duarte WR, Januário AL, Faber J, Bezerra AC. A new method to assess and measure palatal masticatory mucosa by cone-beam computerized tomography. J Clin Periodontol 2009;36:564-8.
Van Assche N, van Steenberghe D, Quirynen M, Jacobs R. Accuracy assessment of computer-assisted flapless implant placement in partial edentulism. J Clin Periodontol 2010;37:398-403.
Chen LC, Lundgren T, Hallström H, Cherel F. Comparison of different methods of assessing alveolar ridge dimensions prior to dental implant placement. J Periodontol 2008;79:401-5.
Suomalainen A, Vehmas T, Kortesniemi M, Robinson S, Peltola J. Accuracy of linear measurements using dental cone beam and conventional multislice computed tomography. Dentomaxillofac Radiol 2008;37:10-7.
Al-Ekrish AA, Ekram M. A comparative study of the accuracy and reliability of multidetector computed tomography and cone beam computed tomography in the assessment of dental implant site dimensions. Dentomaxillofac Radiol 2011;40:67-75.
Kobayashi K, Shimoda S, Nakagawa Y, Yamamoto A. Accuracy in measurement of distance using limited cone-beam computerized tomography. Int J Oral Maxillofac Implants 2004;19:228-31.
Kamburoǧlu K, Kiliç C, Ozen T, Yüksel SP. Measurements of mandibular canal region obtained by cone-beam computed tomography: A cadaveric study. Oral Surg Oral Med Oral Pathol Oral Radiol Endod 2009;107:e34-42.
Berco M, Rigali PH Jr., Miner RM, DeLuca S, Anderson NK, Will LA. Accuracy and reliability of linear cephalometric measurements from cone-beam computed tomography scans of a dry human skull. Am J Orthod Dentofacial Orthop 2009;136:17.e1-9.
Periago DR, Scarfe WC, Moshiri M, Scheetz JP, Silveira AM, Farman AG. Linear accuracy and reliability of cone beam CT derived 3-dimensional images constructed using an orthodontic volumetric rendering program. Angle Orthod 2008;78:387-95.
Brown AA, Scarfe WC, Scheetz JP, Silveira AM, Farman AG. Linear accuracy of cone beam CT derived 3D images. Angle Orthod 2009;79:150-7.
Lagravère MO, Carey J, Toogood RW, Major PW. Three-dimensional accuracy of measurements made with software on cone-beam computed tomography images. Am J Orthod Dentofacial Orthop 2008;134:112-6.
Loubele M, Van Assche N, Carpentier K, Maes F, Jacobs R, van Steenberghe D, et al
. Comparative localized linear accuracy of small-field cone-beam CT and multislice CT for alveolar bone measurements. Oral Surg Oral Med Oral Pathol Oral Radiol Endod 2008;105:512-8.
Sakabe J, Kuroki Y, Fujimaki S, Nakajima I, Honda K. Reproducibility and accuracy of measuring unerupted teeth using limited cone beam X-ray CT. Dentomaxillofac Radiol 2007;36:2-6.
Moreira CR, Sales MA, Lopes PM, Cavalcanti MG. Assessment of linear and angular measurements on three-dimensional cone-beam computed tomographic images. Oral Surg Oral Med Oral Pathol Oral Radiol Endod 2009;108:430-6.
Eshraghi T, McAllister N, McAllister B. Clinical applications of digital 2-D and 3-D radiography for the periodontist. J Evid Based Dent Pract 2012;12:36-45.
Mol A, Balasundaram A.In vitro
cone beam computed tomography imaging of periodontal bone. Dentomaxillofac Radiol 2008;37:319-24.
Scarfe WC, Farman AG, Levin MD, Gane D. Essentials of maxillofacial cone beam computed tomography. Alpha Omegan 2010;103:62-7.
Langer B, Gelb DA, Krutchkoff DJ. Early re-entry procedure. Part II. A five year histologic evaluation. J Periodontol 1981;52:135-9.
[Figure 1], [Figure 2], [Figure 3], [Figure 4]
[Table 1], [Table 2], [Table 3], [Table 4], [Table 5], [Table 6], [Table 7], [Table 8], [Table 9]